Sokratitel channeled is intended for division of tests of bulks into two parts, equivalent on contents.
|Width of fillets, mm||15|
|Quantity of fillets, piece||16|
|Fineness of material of test, mm, no more||5|
|Full volume of probopriyemnik, dm3||8.8|
|Overall dimensions, mm, no more|
|Error of division, % , no more||1|
Maintenance instruction (operating procedure):
The divider of tests is prepared for work: the delitelny funnel is established on support, probopriyemnik draw near closely to each other under a delitelny funnel. A scoop gather a certain part of test and pour out in a delitelny funnel. Process is repeated until it is filled probopriyemnik. Then material from probopriyemnik is poured into separate capacities; empty probopriyemnik establish under a funnel again and continue process of division of test.
If it is necessary to reduce test several times (i. e. to receive 1/4; 1/8; 1/16, etc. of part of test), one part of test received after the first reduction is directed to reduction once again. Thus receive test weighing 1/4 speak rapidly the initial mass of test. Process is repeated until receive the required mass of test.