DKI-N-02 defibrillator (from a network)
Easy-to-work, convenient and reliable, one of the best domestic models of defibrillators that is confirmed by both technical characteristics, and a great demand of medical institutions.
Form of a defibrilliruyushchy impulse — kvazisinusoidny asymmetric bipolar. The impulse gives the chance to reach therapeutic effect of electropulse therapy at energy no more than 190 J that there is twice less energy when using defibrillators with other forms of impulses. It is known that the less energy of a defibrilliruyushchy impulse, the less electric injury of a myocardium and the is more than chances of recovery, especially at reusable defibrillations. For an external defibrillation the fixed doses with energy 45, 65, 90, 140 are provided, 190 J. Terapevtichesky effect are reached at energy, twice smaller from accepted in world practice which considerably reduces probability of damage of a cardiac muscle. The device provides stability of parameters of an impulse, automatic regulation and indexation of pulse current in all range of interelectrode resistance of a body of the patient. It is intended for an electropulse defibrillation of heart through the closed thorax. Can work in the mode of electroanesthesia and cardiosynchronization. Portable, ideally adapted for conditions of the emergency medical care in a hospital and an ambulance car. All governing bodies are taken out on holders of electrodes therefore for use of the device one person suffices.
The fixed doses for a defibrillation. J 45, 65, 90, 140,190
Current amplitude in the first half-cycle at
to loading of 40 Ohms for the maximum dose, A 30 … 50
Ratio between amplitudes of the first and second half-cycles of an impulse 0,5
Supply voltage, In
from the power supply network with a frequency of 50 and 60 Hz 220
from an onboard network of the car 10 … 14,5
Device charging time from a network
220B/from of an onboard network of the car, seconds 10/15
Dimensions, mm 385x342x40
Weight, kg 8
The power supply unit which is used in case of connection of the device to an onboard network of the car of an emergency medical service is included in the package of delivery. The power supply unit is connected to a network of the car and forms tension, necessary for work of a defibrillator. On the panel of the power supply unit there is the corresponding socket.